Statham PA4388-1M, CEC 170087-0104
Has cable over 3.5 ft. long with connector (see picture).
Pins: Input 1+ 4- (28VDC), Output 2+ 3- (5VDC FS), Calibration 5+ 6-, Shield 7.
Dimensions are approx. 1.37 dia x 3.13 inches (body)
Male ¼-inch flared tubing fitting.
Mating electrical connector M83723/86R1407N or MS24266R14T7SN
Shorting the calibration pins produces anelectrical output of 75 percent of full scale (3.75 VDC).
Highly corrosion resistant.
Input voltage 28 + 2 VDC, 75 ma max.
Output voltage 0 psia to full scale = 0.0 to5.0 VDC full scale linear
Response time 4 milliseconds max.
Operating range 30 to 120 °Fplus.
Overall Error (root sum square of all errors) +2 percent of full scale.
Noise 25 millivolts max. peak-to-peak.
Nonlinearity + .05 percent of full scalebest fit straight line.
Hysteresis + .2 percent of full scalemax.
Nonrepeatability +.1 percent of fullscale max.
Thermal sensitivity max. +.02 percentfull scale per degree F from 30 to 150 °F.
Info: Text [phone removed by ].
Paypal, or also accept crypto or Venmo.
Popular TutorialsAlarms and Indicator Circuit diagrams