The full-scale deflection of the universal high-input-resistance voltmeter circuit shown in the figure depends on the function switch position as follows:
(a) 5V dc on position 1
(b) 5V ac rms in position 2
(c) 5V peak ac in position 3
(d) 5V ac peak-to-peak in position 4
The circuit is basically a voltage-to-current converter. The design procedure is as follows:
Calculate RI according to the application from one of the following equations:
(a) dc voltmeter: RIA = full-scale EDC/IFS
(b) rms ac voltmeter (sine wave only): RIB = 0.9 full-scale ERMS/ IFS
(c) Peak reading voltmeter (sine wave only): RIC = 0.636 full-scale EPK/IFS
(d) Peak-to-peak ac voltmeter (sine wave only): RID = 0.318 full-scale EPK-TO-PK / IFS
The term IFS in the above equations refers to meter?s full-scale deflection current rating in amperes.
It must be noted that neither meter resistance nor diode voltage drops affects meter current.
A high-input-resistance op-amp, a bridge rectifier, a microammeter, and a few other discrete components are all that are required to realise this versatile circuit. This circuit can be used for measurement of dc, ac rms, ac peak, or ac peak-to-peak voltage by simply changing the value of the resistor connected between the inverting input terminal of the op-amp and ground. The voltage to be measured is connected to non-inverting input of the op-amp.
5.00V 44 ?A
4.00V 34 ?A
3.00V 24 ?A
2.00V 14 ?A
Position 2 of Function Switch
Erms input Meter Current
Position 3 of Function Switch
EPk input Meter Current
Position 4 of Function Switch